Systematic review of the performance and clinical utility of point of care HIV-1 RNA testing for diagnosis and care

Background Point of-care (POC) HIV-1 RNA tests which are accurate and easy to use with limited infrastructure are needed in resource-limited settings (RLS). We systematically reviewed evidence of POC test performance compared to laboratory-based HIV-1 RNA assays and the potential utility of these tests for diagnosis and care in RLS. Methods Studies published up to July 2018 were identified by a search of PUBMED, EMBASE, Web of Science, CINAHL and Cochrane Central Register of Controlled Trials. Studies evaluating the use of POC HIV-1 RNA testing for early infant diagnosis (EID), acute HIV infection (AHI) diagnosis, or viral load monitoring (VL), compared to centralized testing, were included. Separate search strategies were used for each testing objective. Results 197 abstracts were screened and 34 full-text articles were assessed, of which 32 met inclusion criteria. Thirty studies evaluated performance and diagnostic accuracy of POC tests compared to standard reference tests. Two of the thirty and two additional studies with no comparative testing reported on clinical utility of POC results. Five different POC tests (Cepheid GeneXpert HIV-1 Quantitative and Qualitative assays, Alere q HIV‐1/2 Detect, SAMBA, Liat HIV Quant and Aptima HIV‐1 Quant) were used in 21 studies of VL, 11 of EID and 2 of AHI. POC tests were easy to use, had rapid turnaround times, and comparable accuracy and precision to reference technologies. Sensitivity and specificity were high for EID and AHI but lower for VL. For VL, lower sensitivity was reported for whole blood and dried blood spots compared to plasma samples. Reported error rates for Cepheid GeneXpert Qual (2.0%-5.0%), GeneXpert Quant (2.5%-17.0%) and Alere q HIV‐1/2 Detect (3.1%-11.0%) were higher than in WHO prequalification reports. Most errors resolved with retesting; however, inadequate sample volumes often precluded repeat testing. Only two studies used POC results for clinical management, one for EID and another for VL. POC EID resulted in shorter time-to-result, rapid ART initiation, and better retention in care compared to centralised testing. Conclusions Performance of POC HIV-1 RNA tests is comparable to reference assays, and have potential to improve patient outcomes. Additional studies on implementation in limited-resources settings are needed.


Introduction
Clinical point of care (POC) testing has evolved for situations requiring fast turnaround times and those in which a centralized lab approach faces other limitations, such as difficult or costly transportation of samples [1].POC testing is defined as near-patient testing in a hospital, doctor's office, clinic or home, with the advantage of providing a rapid answer [1]and thus resulting in fewer patients being lost to follow up.Testing occurs while patients are on-site and reduces the burden on patients by circumventing the need for a return visit [2].POC testing can significantly expand access to clinical laboratory testing for rural populations by eliminating the need for sample transport, laboratory and data management infrastructure, and highly trained staff [3].
Many currently available diagnostic assays for human immunodeficiency virus type 1 (HIV-1) infection, including polymerase chain reaction (PCR), enzyme-linked immunosorbent assay (ELISA) and Western blot (WB), are limited to centralized laboratories due to requirements for infrastructure and trained personnel.These laboratory-based assays are complex, expensive and time-consuming, limiting their accessibility in developing countries where the challenge of the HIV/AIDS pandemic is most severe [4].Although rapid antibody tests have greatly expanded access to HIV diagnosis, the inability of these tests to detect HIV-1 RNA and the poor performance of current fourth generation rapid antigen/antibody assays in many high-prevalence settings [5,6] means that resource-limited settings face challenges in the detection of infection in infants and in patients who were recently infected.In addition, HIV-1 RNA quantitation or semi-quantitative viral load (VL) testing is needed to monitor ART [4].POC HIV-1 RNA testing could contribute to timely HIV diagnosis and improve detection of treatment failure, resulting in improved clinical outcomes and reduced HIV transmission [4].
In 2018, Nash et al conducted a review of the performance of the Xpert HIV-1 Viral Load assay, reporting high correlation between POC Xpert results and those of laboratory-based reference assays (pooled Pearson correlation 0.94; pooled Spearman correlation 0.96).Bland-Altman analyses pooled from 11 identified studies were within 0.35 log/copies ml of perfect agreement [11].Numerous field evaluations of other POC HIV-1 RNA assays have since been conducted in different countries and settings.However, to date, there is no systematic review on the performance of all currently available POC HIV-1 RNA testing assays and assessment of their uses in HIV care.
As POC HIV-1 RNA testing has become increasingly available and has been shown to be accurate and valid for multiple clinical uses, this systematic review aimed to synthesize evidence on the performance and clinical utility of POC quantitative (i.e., continuous) or qualitative (i.e., dichotomous) HIV-1 RNA testing assays for different purposes, and to identify barriers and facilitators to their scale up in resource-limited settings.

Eligibility criteria
Studies evaluating the use of POC HIV-1 RNA testing for EID, acute HIV infection (AHI) diagnosis, or VL monitoring, as well as studies comparing the POC HIV-1 RNA testing to centralized HIV-1 RNA testing were included.We excluded studies testing commercially prepared sample panels [12].Publications on laboratory-based HIV-1 RNA diagnostics, POC HIV-2 RNA assays, rapid antibody HIV tests, combined rapid antibody/antigen tests, and POC assays used for diagnosis of infectious diseases other than HIV were also excluded.

Search strategy
The search was carried out in March 2017 and updated in July 2018.PubMed, EMBASE, Web of Science, CINAHL and the Cochrane controlled trial register were searched using the following terms: ("point of care HIV-1 viral load" OR "point-of-care HIV-1 viral load" OR "Xpert HIV-1" OR "GeneXpert" OR Alere OR SAMBA) in combination with (EID OR "early infant diagnosis" OR "infant HIV infection" OR MTCT OR "mother to child transmission") or ("ART monitoring" OR "HIV antiretroviral therapy monitoring "OR "viral load monitoring "OR "HIV-1 viral load monitoring" OR "treatment failure") or ("acute HIV diagnosis" OR "early HIV diagnosis").References of included studies were searched for additional relevant literature.Conference abstracts were excluded due to insufficient detail.

Data extraction
Study titles and abstracts were checked for eligibility according to the inclusion criteria detailed above.Full-text articles were retrieved for potentially eligible studies, and the final set of included studies was agreed upon by all researchers.Data were abstracted by two independent reviewers (CAA and CJN) using a standard data abstraction form that recorded study characteristics, sample characteristics, POC assay evaluated, comparator assay (if any), test accuracy (correlation or concordance, sensitivity, specificity), and error rate.In addition, factors related to barriers or facilitators to the scale-up of POC assays were noted, including those related to human resources, supply chain management, and patient and provider attitudes.

Quality assessment
The quality of the studies comparing POC to reference assays was assessed by two independent reviewers using eighteen of twenty-four criteria selected from the STARD guidelines for reporting diagnostic studies [13]; these criteria were selected based on relevance to the literature reviewed.The eighteen parameters appraised covered six main categories including the title; abstract and key words; introduction; methods (participant eligibility criteria and sampling, reference and index test methods and statistical methods for comparing measures of diagnostic accuracy); results (flow chart of participant sampling, turnaround time to test results for reference and index tests and reported estimates of diagnostic accuracy) and discussion (clinical applicability of the results).Each reviewer scored the publications, with disagreements resolved by discussion with all authors.

Study selection
The search produced 313 references.Following removal of duplicates, 197 titles and abstracts were screened and 34 references were identified for full-text review.An additional 8 studies were identified through searches of the references of included articles.After screening was completed and any discrepancies resolved by the study team, 32 full-text articles met inclusion criteria and were included in the final review.

Characteristics of included studies
Table 1 outlines the characteristics of the 32 included studies.All studies were published between 2014 and 2018.Thirty studies were field reviews comparing the performance of POC HIV-1 RNA tests to a reference standard.Two of the thirty and two additional studies with no comparative testing included reported on the clinical utility of POC results [14][15][16][17].Of the two field reviews that reported on clinical utility [15,17], one was a field feasibility evaluation Abbreviations: ART = antiretroviral therapy, LLD = lower limit of detection, TB = tuberculosis, VCT = voluntary counselling and testing.Measures reported: r = Pearson's correlation, ρ = Spearman's correlation, k = Cohen's kappa coefficient, R 2 = coefficient of determination https://doi.org/10.1371/journal.pone.0218369.t001 study [14] and the other a cluster-randomized trial [16].POC results were used for clinical management in two studies [14,15].The studies enrolled 12,535 infants for EID, 846 adults for AHI screening, 222 adults for HIV detection and 6,975 HIV patients for VL monitoring, and were conducted in sub-Saharan Africa (n = 16 studies, N = 14,744 participants), Europe (n = 5, N = 811), India (n = 3, N = 561), Israel (n = 4, N = 1,633), or multiple sites (n = 4, N = 2,627).
Three studies assessed the performance of POC HIV-1 RNA assays compared to laboratory-based NAAT assays for VL on specimens other than blood plasma [31,34,44].One evaluated VL results using whole blood, plasma, and dry blood spots (DBS) [31]; another whole blood only [34]; and a third used leuko-depleted whole blood (i.e., filtered whole blood) [44].

Acute and chronic HIV diagnosis
Cepheid GeneXpert HIV-1 Qual.Two studies by Michaeli et al [25] and Rakovsky et al [26] evaluated the use of Cepheid GeneXpert HIV Qual for detection of HIV-1 RNA in stored pre-seroconversion samples collected from individuals with recent, documented HIV-1 acquisition.All samples tested reactive on Architect HIV Ag/Ab Combo and Vidas HIV DUO ULTRA 4 th generation assays and had an indeterminate or negative result when tested by the Bio-Rad Geenius HIV-1/2 Confirmatory assay.A high sensitivity (94.9%-100%) and specificity (92.6%-100%) for the detection of HIV-1 RNA was reported in the two studies [25,26].Of note, Cepheid GeneXpert HIV-1 Qual failed to detect HIV-1 RNA in samples from patients with VL suppressed on antiretroviral therapy, as expected [26].
Garrett et al [27] reported on the use of the Xpert Qual for the detection of HIV-1 RNA on whole blood samples from adult known HIV positive women, likely with chronic infection.Of the twenty samples collected, 13 were from patients with detectable viral loads and 7 were virologically suppressed.All except one of the participants (95% sensitivity) with low level viremia (VL 523 copies/ml) were correctly identified by the assay.
SAMBA HIV-1 Qual.Ondiek [24] et al evaluated the use of SAMBA HIV-1 Qual for the detection of HIV using 202 whole blood adult samples of previously known and unknown HIV status, including ART-naïve patients.It was not specified whether participants were acutely or chronically infected with HIV.They reported a sensitivity and specificity of 100% and 99.2% respectively.
Alere q NAT (Quant).One study evaluated the performance of Alere q NAT compared to a reference test in whole blood samples, using several potential thresholds for treatment failure.Sensitivity for identifying treatment failure was 96.8% at 1,000 copies/ml and 84.0% at 10,000 copies/ml; however, specificity was 47.8% at 1,000 copies/ml compared to 90.3% at 10,000 copies/ml [34].Correlation of HIV-1 RNA levels by the two methods used was r 2 = 0.361.
SAMBA HIV Semiquantitative assay.High agreement was reported between the SAMBA HIV Semiquantitative assay and reference tests at a threshold of 1,000 copies/ml, ranging from 96.5%-98.1% [30,40,44].Two studies reported only specificity at a threshold of 1,000 copies/ml, which was 100% in both studies [30,40].
Liat HIV Quant.Scott et al evaluated the performance of Liat HIV Quant compared to reference tests for VL monitoring.A high P c concordance (Pearson's correlation x a bias correlation factor) of 0.96 with the reference test was reported.Sensitivity was 100% at a threshold of 1000 copies/ml for both plasma and whole blood assays, but specificity at this threshold was only 41.2% for whole blood, compared to 88.2% for plasma assays [10].Downward misclassification compared to reference results at the 1000 copies/ml threshold, falsely suggesting virologic suppression, was more frequent with whole blood compared to plasma samples [10].

Error rates
Error rates were reported for four POC assays; Xpert Quant, Xpert Qual, Alere Detect and Liat HIV Quant.Error rates ranged between 2.0%-5.0%for Xpert Qual [14,15,18,23] and between 2.5%-17.0%for Xpert Quant [14,18,31,32,35,39], and were associated with inadequate sample volume ("error" result) [14,15,18,23], incorrect sample processing or PCR inhibition ("invalid" result) [18], insufficient data collection ("no result") [18], mechanical or cartridge errors resulting from faulty modules requiring replacement [15], incorrect pipette supply [15], device optic errors [23], probe check failures [31], and power outages [23].Repeat testing resolved Xpert Qual errors in two studies [15,23], with 10% of errors persisting in another [15].In three studies using Xpert Quant [18,32,35], samples for which errors occurred could not be retested due to lack of extra plasma aliquots.In the study with the highest reported invalid rate (17%) for Xpert Quant [39], the error rate was attributed primarily to cartridges that were broken during shipment or were defective.For Alere Detect, error rates ranged from 3.1%-11.0%[16,17,19,20,23], including operator errors such as no or too little sample being detected, cartridge not properly locked or misaligned, and device errors, such as a connection error between the controller and processor and failure to read the barcode [23].Specimens once retested were reported to produce valid results [23] or resolve the majority of the errors [20].Error rates were observed to be higher in birth testing (infants<7 days) compared to testing in older infants [17,20].For the Liat HIV Quant, a 1.6% error rate was reported, all due to scanning errors [10].

Clinical utility
Four studies reported on clinical utility of POC results [14][15][16][17], and two studies used POC results for clinical management [14,15].Technau et al reported results of POC test implementation within an EID programme in Johannesburg, South Africa.Samples were obtained from HIV-exposed infants for POC testing with the Xpert Qual and compared to the local standard of care, which was HIV PCR testing using the Roche COBAS TaqMan HIV-1 Qualitative test (version 2�0, Roche Molecular Systems, Branchburg, NJ, USA).A positive test by either assay prompted ART initiation and confirmatory testing.A total of 30 neonates were diagnosed using the POC Xpert test, all of whom initiated ART.Time to result was reduced by POC testing (median 1 day vs. 10 days for HIV PCR), leading to more rapid ART initiation in infected neonates identified by the POC test [15].
Ndlovu et al evaluated the operational feasibility of integrated HIV VL, EID and MTB/ RIF testing using the GeneXpert platform [14].At three rural health facilities in Zimbabwe, whole blood samples were collected for HIV VL testing and DBS were collected from infants for EID.POC Xpert EID and VL testing had shorter median turnaround time for result delivery (1 day for each), compared to conventional centralised testing (17 and 26 days, respectively), substantially reducing time to ART initiation and decreasing patient loss to follow-up [14].
Jani et al measured the effect of POC EID on ART initiation rates and retention in care among HIV-positive infants in Mozambique [16].POC EID facilitated rapid diagnosis and treatment of HIV-infected infants: 89.7% of HIV positive infants in the POC arm and 12.8% in the standard of care arm initiated ART within 60 days of sample collection.At 90 days of follow-up, 61.6% of those who initiated ART in POC arm and 42.9% in the standard of care arm were retained in care [16].
Meggi et al evaluated the feasibility, performance and diagnostic yield of rapid POC EID at birth within primary health care maternity wards in Mozambique [17].Samples obtained from HIV-exposed infants were tested at birth and at 4-6 weeks using both POC Alere q HIV-1/2 Detect Qual and Roche COBAS TaqMan HIV-1 Qualitative test (version 2�0).Sensitivity and specificity of POC testing compared with laboratory testing at birth were 100% (95% CI 89.4 ±100.0) and 100% (95% CI 99.8±100.0),respectively.Notable within the study were results of four infants who tested positive for HIV infection with laboratory-based and/or POC EID nucleic acid tests at birth but tested negative at least once during follow-up while on nevirapine (NVP) prophylaxis.For all four infants, results turned positive following NVP cessation, indicating that the diagnosis at birth was correct [17].

Provider experiences with POC assays
Cepheid GeneXpert HIV-1 Qual and HIV-1 Quant.Table 2 outlines the provider experiences with POC assays as obtained from the included studies.
These assays were found to be simple to use [32,35,36,43], with a rapid turnaround time of 90 minutes [18,32,36,43], resulting in expeditious clinical decision-making [43].Other advantages included: results that were easy to interpret [23], more efficient patient management compared to batch testing in a central laboratory [43], no requirement for calibrated mechanical pipettes, and the ability for less skilled personnel (potentially including non-laboratory personnel) to perform the test [15,23,35].The GeneXpert platform was found to be compact in comparison to other platforms and more cost effective (�17 USD per cartridge) [21,36,43].The machine that takes a 4-cartridge module is easy to transport for use in various settings, an advantage for low-income countries [32].The ability to run assays for tuberculosis and other pathogens in addition to HIV-1 was also advantageous [14,18,31,36,38], allowing for use of the same procurement chains and service plans for both TB and HIV diagnostics [18].Of note, the Cepheid GeneXpert platform can be used in large laboratories with high volume, as configurations for up to 80 modules are available [31].The modular nature of the equipment makes it possible to change one module if not working, avoiding a complete halt in the platform activities [18].Drawbacks of the Xpert assays included the need for air-conditioning and refrigeration to maintain the platform and samples at correct temperatures [14] and the requirement for a back-up power source for the high-throughput modules when a steady power supply is not readily available [23].In addition, overloading may occur when a large volume of tests is performed on a single machine and quality control requirements may exceed local capacity [14,27].Where implementation is planned for centralized testing, efficient and robust sample transport networks are needed [14].Cartridge colours for the different Xpert assays are all the same, and therefore separate preparation areas in the laboratory are required to avoid confusion [14,31].Samples must also be manually added to the cartridge [23,32].Centrifugation is required to obtain plasma samples [14,31,38], and the 1000-μl plasma volume required could be a particular challenge for testing infants [38].Finally, one study reported an instance of contamination of a DBS sample with a plasma VL sample, generating a false positive EID result and highlighting the risk of contamination with PCR assays [14].
Alere q HIV-1/2 Detect Qual.Advantages of this assay include: ease of use with a short run time of about 52 minutes [23], limited training required [16,17,19,22], a small sample volume requirement (25μl) [19], the ability to print out test results, and the availability of a dedicated battery pack as an alternative backup power source [23].Disadvantages include the ability to run only one sample at a time and difficulty using capillaries to load samples [23].The cartridge ejection mechanism was reported to cause problems and the keyboard design was not considered user-friendly.The lack of information provided when an error message occurred was reported as a further disadvantage in one study; however, valid results were obtained on re-running the specimen [19].
SAMBA HIV Semiquantitative assay.The short turnaround time of 90 minutes, heat stability, and limited training required have been reported as advantages [30,40].The visual detection of nucleic acid on a test strip was an added advantage, enabling staff to show result to the patient [40].Limitations reported include inability to provide a quantitative result and requirement for a plasma sample, with no potential to use DBS or other sample types [30].Despite a stronger signal on the test strip distinguishing positive results from negative results than present in current HIV rapid antibody tests, the possibility for misinterpretation of results by users remains [40,44].This issue can be overcome by having a paper printout of the test result, or having the results appear on a screen [40].
Liat HIV Quant.This assay was found easy to use, with no support required for installation and only a brief 2-hour training needed [10].The closed system requires no special safety precautions.Data can be exported from the device using a USB port or ethernet cable.The Liat platform can test 12-14 specimens in an 8-hour day, using either whole blood or plasma.Disadvantages reported include the short half-life of reagents, the need to maintain a cold chain for the testing cartridges, and the need for centrifugation when plasma is used [10].
Aptima HIV-1 Quant Dx Assay.The reported advantage of this assay is its ability to detect low HIV-1 copy numbers [33].

Quality assessment
Quality was assessed based on the 18 modified STARD criteria for the 30 included studies that reported on diagnostic accuracy (Table 3).Most (86.7%, n = 26) were easily classified as studies of diagnostic accuracy, with a clear objective of comparing the diagnostic accuracy of a POC assay to one or more laboratory-based reference assays.All studies stated whether or not their results were applied clinically.Few studies, however, specified inclusion or exclusion criteria (33.3%, n = 10), clinical and demographic data for the population from which samples were obtained (36.6%, n = 11), data collection procedures (16.7%, n = 5), or training requirements for the POC assay evaluated (26.7%, n = 8).

Discussion
This systematic review of the literature aimed to synthesize evidence on the performance and clinical utility of POC quantitative or qualitative HIV-1 RNA testing assays for EID, AHI diagnosis and VL monitoring, and to identify barriers and facilitators to their scale-up in resourcelimited settings.We found 32 studies that met inclusion criteria, of which 30 focused on diagnostic accuracy and 4 included results on clinical utility.Overall, the studies of diagnostic accuracy showed excellent performance.Where correlation between quantitative results was reported, it was high across all assays assessed; in addition, inter-assay differences reported were <0.5 log copies/ml, which is considered for clinical practice [45,46].Although POC VL assays tended to overestimate virologic failure compared to reference tests, which could lead to early switching, the use of central laboratory testing to confirm treatment failure could overcome this disadvantage [48].In addition, while sensitivity for birth testing in infants <7 days of age was lower than for routine EID testing at 6-14 weeks, earlier diagnosis for those infants with positive POC results provides important advantages [14,16].Given the increasing body of evidence on the diagnostic accuracy of POC RNA testing, studies on clinical utility, implementation barriers and facilitators, and cost-effectiveness should be the focus of future research.
The WHO recommends HIV-1 testing of exposed infants at the earliest opportunity with an assay that detects HIV-1 DNA or RNA, by 4-6 weeks of age at the latest [49].Without ART, about 50% of perinatally-infected infants progress to advanced disease by 8-12 weeks of age or die [50,51].Unfortunately, only half of all HIV-exposed infants in RLS receive an EID test within 2 months of age [49].Challenges with current EID programs include loss to followup of 30%-80% of mother-infant pairs and late presentation of many infants, who miss out on the benefits of early ART initiation as well as lack of diversified testing and sample collection sites [49,52].In the two studies that used POC EID results, turnaround time was reduced to one day, leading to rapid ART initiation and reduced loss to follow-up [14,15].With the scaleup of prevention of mother to child transmission (PMTCT) programmes in sub-Saharan Africa, where all EID studies in this review were conducted, POC testing by non-specialized personnel in field settings could help decentralize services and improve infant outcomes [19].Integration of POC HIV testing services for women and infants in other high-yield settings for paediatric HIV case-finding, including TB clinics, malnutrition clinics and inpatient wards, could lead to the identification of HIV-exposed or -infected infants missed by routine PMTCT programmes [52].While effects of infant NVP prophylaxis on birth testing outcomes and algorithms for confirmatory testing before ART initiation still need to be addressed [17], POC testing for EID holds promise.
The WHO estimates that by mid-2016, more than 18 million HIV-infected individuals were receiving ART, with access increasing due to "test and treat" approaches [53,54].With continued scale-up, virologic monitoring to ensure treatment efficacy and combat HIV drug resistance is necessary [53,54], particularly in low-and middle-income countries, where delayed treatment and poor care engagement can result from financial, human resource, and infrastructural barriers [55,56].Currently, only �20% of ART patients in low-middle income countries receive VL testing [54,57,58].This low coverage has been attributed to a number of challenges, including poor sample referral systems, a lack of electronic data systems for results, and long turnaround times resulting in patient loss to follow-up [56].Improved health data systems are needed to flag those in need of VL testing and ensure fast turnaround times for prompt clinical decision-making [56,58].POC VL testing could meet a critical need, by ensuring same-day results for providers in rural or hard-to-reach areas where VL test access is currently limited [57,58].The advantages of POC VL assays may outweigh concerns about upward misclassification, especially as the current tendency is to switch too late, rather than too early [56].Centralized confirmatory testing with standard-of-care assays for patients with suspected virological failure could address concerns about false positive results.AHI diagnosis is a concern that has often been overlooked in RLS [59].An estimated 10%-50% of all HIV transmission events may be attributable to AHI, a period associated with high transmission risk due to extremely high viral load and high infectivity of founder viruses [60].Despite the importance of diagnosing AHI, there is currently no WHO recommendation on AHI diagnosis, and very few studies have considered this application of POC HIV-1 RNA assays [48].New WHO recommendations to exclude acute or early HIV infection prior to initiating PrEP (pre-exposure prophylaxis) or PEP (post-exposure prophylaxis) [48] may provide the impetus needed to investigate the utility of POC HIV-1 RNA assays for AHI diagnosis among patients with symptoms of acute retroviral syndrome.In our review, we found two studies investigating performance of the Cepheid GeneXpert HIV-1 Qual for AHI diagnosis.Additional studies are on the horizon, including the Tambua Mapema Plus study, a proof-ofconcept study evaluating the impact of an HIV-1 RNA testing intervention (Cepheid GeneXpert HIV-1 Qual) targeting young adult Kenyan patients aged 18-39 years who seek urgent care for symptoms associated with AHI (ClinicalTrials.govIdentifier: NCT03508908) [61].Prompt diagnosis of AHI is needed to maximize the benefits of a test and treat approach [59].If AHI diagnosis is augmented by assisted partner notification with HIV-1 RNA testing, partners with acute or prevalent HIV infection could also be identified and linked to ART if infected or PrEP if uninfected, maximizing impact.
Various factors should be considered prior to implementation and scale-up of POC HIV-1 RNA testing.Staff will require training and rigorous quality control measures should be put in place.Costs for reagents and consumables, shipment, customs charges, tax, service, and maintenance remain important considerations, which we were unable to address in the current review given the paucity of published data on this aspect of POC HIV-1 RNA tests.However, 2017 data from the Global Fund suggests the maximum price per test including consumables is $10.60 per test for the Aptima Quant Dx assay for VL, $16.80 per test for Xpert Quant for VL, $17.95 per tests for Xpert Qual for EID, up to $25 per test for Alere Q for EID, and $37.40 each for SAMBA I and II for VL and EID [62].The cost of the POC-of-care equipment excluding service and maintenance varies greatly; from $12,280 for the GeneXpert IV-2 platform, $24,800 for SAMBA II, and $25,000 for Alere Q, up to $71,500 for the GeneXpert XVI and $72,000 for SAMBA I, and as high as $150,000 for the Hologic panther system on which the Aptima Quant Dx assay operates [62].The primary strategy proposed to mitigate cost constraints is pooled procurement through PEPFAR or the Global Fund, with the hope that eventually higher demand will bring about competition and drive down costs [57].
Our review has a number of limitations.First, only published studies, of which the majority were field reviews, were included.Very few studies reported on clinical utility of the POC devices.Second, conference abstracts were excluded due to insufficient detail and inability to assess the quality of the study.This resulted in the exclusion of some relevant studies.Third, pricing data are often difficult to find and can change rapidly.Lastly, there was variability in the measures of accuracy, precision and agreement reported across the studies, which did not allow for pooling of data and limits its generalisability.

Conclusion
This systematic review has identified a number of studies investigating POC HIV-1 RNA assays in RLS, of which most demonstrate acceptable clinical accuracy.Very few studies have investigated clinical utility and strategies for scale-up.As POC HIV-1 RNA assays are more widely evaluated for the uses discussed in this review, the requirement for plasma samples and thus continued need for trained phlebotomists and centrifugation of samples may remain a barrier in some RLS [57].In addition, the need for additional resources such as air-conditioning, cold chain for reagents and a back-up power source may prove an additional challenge.If POC HIV testing is considered as part of an integrated laboratory network, strong tracking systems, good documentation, and robust sample transport and supply chain systems are needed.Moving forward, further research is needed on clinical utility, quality assurance, algorithms for confirming positive results, reliability of results for clinical decision-making, and cost-effectiveness [63].In general, however, it is clear that POC HIV-1 RNA assays are here to stay and offer clear advantages that will help advance HIV prevention and care globally.

Fig 1
details the study selection flow chart.

Fig 1 .
Fig 1. Flow diagram of study selection.https://doi.org/10.1371/journal.pone.0218369.g001 Fig 2 presents a Forest Plot of the sensitivity of Cepheid GeneXpert Qual assay compared to reference tests for EID.
Fig 3 presents a Forest Plot of the sensitivity of Alere Detect compared to reference tests for EID.